Uniquely designed for greater comfort. Use only with Accu-Chek Softclix and Accu-Chek Softclix Plus lancing devices. Our Accu-Chek Customer Care Service Center is available to answer your questions 24 hours a day. Call 1-800-858-8072. Manufactured in Japan.
Contains: Do not use if protective guard has been previously removed form lancet. Do not re-use lancets or store lancets in device. Caution: This notice reminds the purchaser that a restricted patent license prohibits the use of the Accu-Chek Softclix and Accu-Chek Softclix Plus lancing devices with any lancet other than Accu-Check Softclix lancets.
1. Remove the cap from the lancing device. 2. Insert a new lancet in the lancet holder. It is properly positioned when it snaps into place. To remove the protective cap, twist and pull it away from the lancet. 3. Replace the cap onto the lancing device. Twist the cap of the lancing device to set the lancet depth. Start at 2 or 3. For tougher skin, dial to a higher number. 4. Press the plunger on the end of the device all the way down until it clicks. A yellow dot will appear and remain in the center of the clear release button. 5. Wash your hands in warm, soapy water. Dry well. 6. Hold the lancing device against the side of your fingertip. 7. Press the release button. Gently squeeze the area until you get a small sample of blood. 8. Apply the drop of blood to the target area of your test strip. 9. Immediately remove used lancet. To avoid accidental injury, do not store lancet in device. Attention: Please read the Accu-Chek Softclix lancing device package insert for product warranty and other information before use.
About the Producer